The lab x-ray source situated at the end of the XMaS beamline is based on a copper microfocus tube (Vari Aqua 0.080NWCMF24 Petrick GmbH, Germany). The electron beam spot size at the water cooled anode is approximately 50 microns, with an electron beam energy of 40 keV. An INEL XRC60 power supply is used to power the tube, which contains electronics to drive the x-ray shutter, that is mounted within the tube. Also, the power supply houses the on board water cooling system to dissipate the heat generated at the anode.
The power supply and shutter can be controlled remotely by using a Linux window when all of the appropriate interlocks are in place. All of the safety interlocks are based on the ESRF personal safety system and the appropriate online training must have been followed to use the source. The operating system of the offline x-ray source is based on the SPEC control system (www.certif.com), as used on the beamline.
Window used to control x-ray tube and shutter on XMaS lab source.
The Collimating Mirror
Schematic of Mirror Geometry
The Germanium <220> Channel Cut Monochromator.
For high resolution work, there is the option to install a Germanium <220> channel cut monochromator into the beam. This allows the user to separate out the copper Kα1 and Kα2 lines and have a very monochromatic beam. However, intensity is reduced by approximately a factor of 20. For normal use, the Germanium monochromator is tuned to the Kα1 line.
Separation of the Copper Kα1 and Kα2 lines using the Germanium monochromator.
A Huber 4 circle diffractometer is available for sample mounting. This instrument has a nearly identical Eulerian cradle as the main instrument on the synchrotron beamline and most of the sample environments available on the XMaS beamline are also compatible with the offline x-ray source.
(1) M. Montel - "The X-Ray Microscope with Catamegonic Roof-Shaped Objective" in: X-ray Microscopy and Microradiography, Vol.5, 1957, pp 177 - 185